About this product
Part Number: CT-27009-31D
Premium quality, ULTRA-narrowband 3nm SII Filter
Chroma SII filters isolate the spectral emission of singly ionised sulfur atoms, also common in nebulae. Emission is at two lines occurring as a doublet, at 671.6 and 673.1 nm. This filter also separates sulfur emission from H-alpha emission (656.3nm).
Chroma Technology is an employee-owned company based in Vermont, USA. Chroma manufacture the highest quality astronomy filters with durable, sputtered hard coatings using single substrates of the best glass, eliminating the need for laminations. All primary filter coatings are applied on the front surface and anti-reflection coatings on the rear surface to prevent ghosting and to maximise transmission. Chroma astronomical filters are used worldwide in research and amateur applications.
Highly precise and accurate, the passbands of these filters remain spectrally stable and do not drift in response to extreme temperature fluctuations or changes in humidity.
All filters may be used with focal ratios of f/4 or slower. For optics faster than f/4 then due to the spectral shift we recommend using the 5nm filters instead.
- CWL: 672.4nm, Bandwidth: 3nm
- Durable sputter coatings
- Parallelism: <30 arcsec
- Thickness: 3.0 +/-0.05mm
- Filters designed for use with CCD and for f/4.0 or slower
- Transmitted wavefront better than 0.25 waves/inch
- No reflections leading to image distortions or ‘back reflections’
Cleaning and Handling
Handle coated pieces by the edges only. Clean gently only if necessary. Loose particles should be removed with a bulb puffer or filtered, pressurised air cleaner. If necessary, gently wipe surface using anhydrous alcohol and lint-free lab towels. Use new surface of towel with each wipe.
- AVOID TOUCHING OR WIPING A/R COATED OR METAL MIRROR SURFACES
- AVOID HANDLING EXPOSED COATINGS WITH BARE FINGERS
Proper orientation of the filter is necessary in order to minimise autofluorescence and maximise performance. There is a caret (arrow) located on the edge of each filter in order to aid orientation. Excitation (x) filters should be positioned with the arrow pointing toward the specimen, toward the inside of the cube, and away from the light source. Emission (m) filters should be placed with the arrow pointing toward the specimen, toward the inside of the cube, and away from the detector/eye. Dichroic mirrors should be mounted with the coated surface toward the light source, excitation filters, and the specimen. The dichroics either have an arrow on the side pointing to the coated side, or they are beveled on the coated side. The beveled side is the smaller surface.
|Average Rating (2 Reviews): |
Monday, 30 March 2020 | Tareq
I received the filter safely, it is as expected, nice quality filter, but the only downside about it is the price and the shipping was a lot and then the VAT also added a lot, so i paid too much for this filter almost close to Astrodon which i try to avoid, DHL always charge more, and that is why i gave it 3 stars, but the quality is 5 stars no doubt.
Thursday, 26 March 2020 | Thomas
I just got to use these 3nm filters (set). My original filters were Baarder OIII8.5 ,ha7 & zwo sii7 which is not bad, but these filter are in another league. Iím very happy with my purchase 🙂
Please fill in the form below to ask a question.