Home > Filters > Imaging & Photography > Narrowband
(Details shown apply to the default selection)
Model: chroma_low_glow_125Part Number: CT-27030-1.25M
Handle coated pieces by the edges only. Clean gently only if necessary. Loose particles should be removed with a bulb puffer or filtered, pressurised air cleaner. If necessary, gently wipe surface using anhydrous alcohol and lint-free lab towels. Use new surface of towel with each wipe.
Proper orientation of the filter is necessary in order to minimise autofluorescence and maximise performance. There is a caret (arrow) located on the edge of each filter in order to aid orientation. Excitation (x) filters should be positioned with the arrow pointing toward the specimen, toward the inside of the cube, and away from the light source. Emission (m) filters should be placed with the arrow pointing toward the specimen, toward the inside of the cube, and away from the detector/eye. Dichroic mirrors should be mounted with the coated surface toward the light source, excitation filters, and the specimen. The dichroics either have an arrow on the side pointing to the coated side, or they are beveled on the coated side. The beveled side is the smaller surface.
Be the first to Write a Review for this item!
First Light Optics LtdUnit 7 Budlake UnitsBudlake RoadMarsh Barton Trading EstateExeter, DevonEX2 8PY
Company No. 5892293VAT No. 907 2895 01
First Light Optics Ltd All rights reserved © 2023